Single-point calibration of the RTS node is easy to perform. One method is presented here using the water ice-point and works with both waterproof and non-waterproof temperature sensors. The RTS firmware provides both offset and gain calibration coefficients. For most purposes, a single point offset calibration is probably sufficient. To accurately perform both gain and offset calibration requires more sophisticated VBE equipment and is beyond the scope of this investigation.